Tape Flux Measurement Revisited
Analog magnetic recording standards specify recorded signal magnitude in terms of short-circuit flux per unit track width, called fluxivity. Two flux measurement methods are standardized: a direct ac method using a high-efficiency head (AES7/ANSI Standard S4.6); and a transfer-to-dc method using a fluxmeter (German Standard DIN 45 520).: A transfer-to-dc measurement made in the 1950s is used to calibrate the fluxivity of German calibration tapes. When these tapes are measured by the direct ac method, the fluxivity is 10% less than the stated value. Thus the 1950s German transfer-to-dc measurement was 10% more sensitive than the direct ac measurement.: A new comparison of the two measurement methods shows that they actually have the same sensitivity within an experimental error of about 0.5%. Thus we conclude that the German measurement in the 1950s must have had a 10% error, and that the fluxivity on all of the calibration tapes based on the old German measurement is 10% less than the stated value tapes identified as 320 nWb/m are actually 288 nWb/m.
Click to purchase paper as a non-member or login as an AES member. If your company or school subscribes to the E-Library then switch to the institutional version. If you are not an AES member and would like to subscribe to the E-Library then Join the AES!
This paper costs $33 for non-members and is temporarily free for AES members.