An electrostatic transducer, having a high internal acoustic impedance and fed by rectangular voltage pulses, is the source of acoustic volume velocity transmitted to the measured acoustic impedance Z/ax. The spectrum of the sound pressure pulses p(t) is obtained with an FFT processor. Measurements are taken first on a cylindrical waveguide and then on the unknown Z/ax. the logarithmic difference of the two curves gives the normalized value of Z/ax. Examples are shown for measurements of acoustic impedance and acoustic transfer impedance.
Click to purchase paper as a non-member or login as an AES member. If your company or school subscribes to the E-Library then switch to the institutional version. If you are not an AES member and would like to subscribe to the E-Library then Join the AES!
This paper costs $33 for non-members and is free for AES members and E-Library subscribers.